Notes on the Jeti Specbo Spectro and Plasma Sets
- This topic has 5 replies, 4 voices, and was last updated 10 years ago by .
Yes, this is annoying, but hope some of this helps.
Seems the Jeti has issues reading bright patterns off plasma TVs. The readings either take an unusual length of time or causes the program to lock up. Here are some solutions …
In both the THX CP program and now the Calman 4.2.2 program (build 232), if you are using a Jeti, you will have to tell the program what type of display you are using. Spec60 on the CP program and Plasma 60 on Calman.
But the goodness does not quite end there though. 🙁 Setting it for 60 does not always guarantee that this is the best choice. If you have it set for 60 and still find it locking up or taking too long to take a reading of 100% … then try setting the probe to just Spec … (not spec60 or spec50) and it might work better.
Also watch out for repeatability issues at the same bright end. If you are having a lot of variation, then try the other setting. You won’t know until you try it.
Some Field Data … LG plasma … 2010 model … spec 60 setting does not work right … try spec …
Samsung Plasma 2009 model … spec does not work … but spec 60 works and results are repeatable.
- You must be logged in to reply to this topic.