You Know Your Meter Needs to be Re-Certified When……
- This topic has 7 replies, 3 voices, and was last updated 9 years, 10 months ago by .
I have an Eye One Pro with Rev D. What are the symptoms when your meter needs recertification?
I ask because I have been getting inconsistent results using Calman 4.3. I did a test with the Pro Workflow. I ran through the pre cal layout about 5-6 times over the course of a few hours. I re-initialized the meter before each run. The WB dE grew over the course of time from about 11 to 18. The gamut dE grew as well from about 12 to 16.
Then I used the greyscale layout in Pro Workflow and took 4-5 more reads and the dE stayed at about 17 plus or minus 1.
At this point, I don’t know what to trust. Any thoughts?
- You must be logged in to reply to this topic.